Inspect quick disconnect coupling half for
KEY to figure 10-1.
damaged cap, nose piece, or clip. Check that cable
assembly is not damaged, broken or missing.
Ejector assembly support
Repair and Replacement.
Quick disconnect coupling half
Replace parts if threads are damaged
beyond simple repair.
Repair tube by welding; replace loose
rivets. If ejector tube is damaged beyond simple repair,
replace the tube.
11. Cable assembly
Replace damaged or missing quick
disconnect coupling half components and replace seal.
14. Ejector tube
Replace packing (13).
Loosen nuts (12) and remove the quick
disconnect coupling half (5), nuts (12), and packing (13).
Reassembly. Refer to figure 10-1 and
reassemble the inflation and deflation ejector assembly
(3) Disassemble the quick disconnect coupling
by reversing the disassembly sequence.
half by removing the cap (6) and seal (7) from the nose
piece (8). Leave cable assembly ( 1) and sleeve (10)
attached unless repair is necessary. Remove clip (9) by
10-3. Box and Valve Manifold Assembly
sliding it off opposite end of nose piece from the cap.
The box and valve manifold
assembly connects the inflation and deflation ejector
c. Cleaning and Inspection
assembly to the MUST hospital inflatable element. Its
Clean metal components of the assembly
main components are the junction box, the two
with solvent and dry thoroughly with compressed air.
regulating valves, and the manifold cap.
Inspect threaded parts for damaged,
stripped or crossed threads.
and washers (2, 3) to remove regulating valve (4).
Remove manifold cap (5) from junction box (6).
(3) Inspect ejector tube for cracks, bends, and
broken welds. Check that wire mesh in the tube is not
Cleaning, Inspection, and Testing.
torn or clogged. Check for loose rivets.
Clean the metal parts with solvent
and dry thoroughly.